Number of views:
VHP Sterilizer Pass Box
Number of views:
Product serial number
The VHPB series sterilization transfer cabin is a special sterilization equipment that can provide absolute biological decontamination on the surface of the article when transferring articles between different functions. It is equipped with a dedicated hydrogen peroxide VHP generator, a sterile air supply system, an electromagnetic door interlocking system, a closed system, a residue removal system after sterilization, a dedicated HMI (human machine interface) and a sterilizing medium delivery system. It is widely used for surface sterilization at room temperature in places such as pharmacy, medical treatment, sanitation, biological experiment, etc.
3. Brief description of work process
The working process is divided into five stages: pre-sterilization cabin airtightness detection, system preheating and dehumidification, microbial decontamination and sterilization, residue removal, and Class A laminar flow ventilation.
3.1 Check whether the airtightness of the chamber can meet the sterilization function before sterilization and whether the airtightness meets the requirements.
System leakage rate detection-inflate the sterilization chamber to make the pressure in the chamber reach the set value, maintain the pressure, and the leakage amount within the set time is not greater than the specified value, indicating that the system is tightly sealed and can enter the sterilization process.
3.2 Preheating and dehumidification of the system-sending clean and dry air into the cabin to keep the humidity in the cabin at a low value, which is conducive to hydrogen peroxide sterilization.
3.3 Sterilization procedure-input hydrogen peroxide vapor to the sterilization chamber, and maintain the expected concentration of the sterilization chamber, keep the VHP concentration above the set concentration, and maintain the sterilization time for more than 30-60 minutes (chemical test paper and biological Test paper verification);
3.4 Residue removal-stop the input of hydrogen peroxide vapor, switch the sterilization system to the residue removal system, so that the hydrogen peroxide gas in the sterilization chamber circulates through the catalyst for decomposition, quickly reduces the hydrogen peroxide concentration, and then ventilates to achieve Hydrogen peroxide remains below 1PPM.
3.5 Class A laminar flow-the air supply fan is turned on, and the fresh air valve and exhaust valve are adjusted to keep the pressure difference between the cabin and the high-level cabin greater than 10Pa. The air flow can be horizontal unidirectional flow or vertical unidirectional flow. After the sterilization and removal of residues are completed, the cabin maintains a laminar flow state. After the high-level side door is opened, the cabin is still in a laminar flow air supply state.
4. Main parameters
Material: External 304 stainless steel, internal cavity: 316L stainless steel
Cleanliness class CLASS A
Leakage rate ≤0.5%VOL/h (under 100Pa)
Air change ≧80T/h
Vaporization temperature ≤60℃
Compressed air ≥0.5Mpa
Sterilizer better than 30% food grade hydrogen peroxide solution
VHP residue removal efficiency ≤1PPM
Control method manual or automatic
VHP pass box
Scan the QR code to read on your phone
We could not find any corresponding parameters, please add them to the properties table
Time of issue:2020-02-20 00:00:00
SUZHOU HJCLEAN TECH Co., LTD
Address: 18,Lou Feng Chuangtou Industrial Area,Suzhou Industrial Park,China
Explanation of Enquiry
Method of contact and occupation